- Open Access
Reduced dose iterative reconstruction versus standard dose filtered back projection in detection of bladder tumors
Egyptian Journal of Radiology and Nuclear Medicine volume 51, Article number: 100 (2020)
The purpose was to assess radiation dose, image quality, and diagnostic performance of reduced-dose scanning with iterative reconstruction (IR) compared with standard-dose with filtered back projection (FBP) with CT urography for detection of bladder tumor. This study was prospectively conducted on 21 patients with bladder masses. All patients were subjected to two scanning protocols: protocol A (standard dose with FBP) and protocol B (additional limited scan to the pelvis at delayed phase with low dose with IR). Based on body weight (< or > 80 kg), each protocol was subdivided into 2 protocols A1 (130 kVp) and A2 (130 kVp) and protocols B1 (80 kVp) and B2 (110 kVp). Radiation dose was assessed in terms of mean CT dose index (CTDI), Dose-length product (DLP) and effective dose (ED). Image quality and diagnostic accuracy were compared in both groups.
Mean CTDI, DLP and ED were reduced by average 72.3 % in the 80 kVp protocol (B1) and by 36.3% in 110 kVp (B2) protocol compared to standard-dose protocols. There were significantly lower SNR (signal to noise ratio) between protocol A1 and B1 at aorta and psoas muscles. Subjective image quality analysis revealed no statistically significant differences between the protocol A2 and B2 whereas there were significant differences between protocol A1and B1 as regards to visual image noise and overall image quality. Diagnostic accuracy was identical among different protocols.
CT urography with IR scanning showed reduced radiation dose and no difference in detection of urothelial carcinomas from standard dose with FBP despite of degraded image quality in 80 kVp scanning.
CT Urography (CTU) is one-step technique that allows full evaluation of the urinary tract for urothelial tumors, renal masses and stones in a single examination. Meanwhile, in traditional CTU protocols patients receive radiation dose of about 54 mGy, the reduction of radiation dose in CTU has been an important item to be resolved [1, 2]. Radiation dose reduction with filtered back projection (FBP) reconstruction techniques is associated with increasing image noises and streak artifacts . With general technical evolution, these drawbacks of FBP have driven the rise of iterative reconstruction (IR) algorithms. The current literature shows that clinical implementation of IR into CT protocols provides essential promise for major improvements in image quality, chiefly noise reduction together with subsequent radiation dose reduction and artifact suppression . The purpose of this study is to compare radiation dose, image quality and diagnostic performance of CTU using reduced-dose scanning with IR versus standard-dose with FBP for detection of urinary bladder tumors.
Twenty-one consecutive patients presented to Alexandria University Hospitals with sonographically diagnosed urinary bladder masses were enrolled in the study between August 2016 and December 2017.
Enrolled patients had serum glomerular filtration rate greater than 60 ml/min/1.73 m2. Patients with contraindications to the use of iodinated contrast media or hip prosthesis were excluded. All patients included in the study were subjected to full history taking including personal data and clinical examination and Routine laboratory investigations.
CT data acquisition
All patients were subjected to two scanning protocols:
Protocol A: standard dose with FBP
Protocol B: additional limited scan to the pelvis at delayed phase with low dose with IR.
Each protocol is subdivided according to body weight into two protocols:
A1 (130 kVp) and A2 (130 kVp) and protocols B1 (80 kVp) and B2 (110 kVp) based on body weight.
Patients with body weight less than 80 kg (n = 10) to standard dose with FBP 130 kVp (protocol A1) and additional scan limited to urinary bladder at delayed phase with low dose with IR 80 kVp (protocol B1) while patients with more than 80 kg (n = 11) into standard dose 130 kvp (protocol A2) and additional scan limited to urinary bladder at delayed phase with (protocol B2); low dose 110 kvp.
All CT examinations were performed with SOMATOM Perspective 64 MDCT (Siemens Healthcare, Forchheim, Germany) at Alexandria Main University Hospital.
Image acquisition parameters were detector collimation 1.2, slice thickness 1.5 mm, field of view 51 cm, rotation time 0.6 s, helical pitch 0.6, tube current was automatically modulated by the machine, and ranged from 165 to 200 mAs. Additional scan limited to urinary bladder was acquired at delayed phase with scan parameters identical to that of standard dose except for the tube voltage which was reduced to 80 kVp (protocol B1) or 110 kVp (protocol B2) according to body weight.
The injection protocols were the same in all examination, 100 ml of iohexol (Omnipaque 300 mg/ml; GE Healthcare Ireland, Cork, Ireland) which was administered into antecubital vein with flow rate of 2–4 ml/s via a pump injector (Medrad mark V, Bayer healthcare, Pennsylvania, USA). Patients consumed 600 ml water orally while waiting for 1 hour before undergoing CT to allow distention of the urinary bladder. Images were obtained at unenhanced, nephrographic (90–110 s after administration of IV contrast medium), and delayed (420 s after administration of contrast medium) phases for standard FBP protocols, scan range started from above the dome of diaphragm to ischial bones as determined from the AP topogram. Delayed phase only for low-dose protocols limited to the urinary bladder as determined from the prior scans.
Raw data of reduced-dose CT examinations in the study group were reconstructed using both mathematical algorithms, FBP and IR (SAFIRE). SAFIRE has 5 preset strength levels. According to recommendations by the vendor, a medium strength level of 3 was applied to all reconstructions in our study. Reconstruction parameters included a B20s + smooth convolution kernel for FBP and the equivalent I30/3+ medium smooth kernel for IR.
Radiation dose report (for the pelvis)
For each patient, volume CT dose index (CTDI) and the dose-length product (DLP) were recorded from the automatically generated patient protocol. The effective radiation dose was then estimated in millisieverts (mSv) (the dose-length product multiplied by the conversion factor of 0.015)  for each scan.
Both objective and subjective image quality were assessed for all patients in a blinded and randomized fashion using Syngo via Workplace; Siemens Healthcare.
Objective image quality
The CT attenuation value (in Hounsfield units HU) of aorta and psoas muscle, image noise (standard deviation of CT attenuation values SD), signal-to-noise ratio (SNR), and contrast-to-noise ratio (CNR) were assessed within the identical two areas (psoas muscle and abdominal aorta) by using a region of interest (ROI) of at least 40 mm2 at level of L4 before bifurcation of abdominal aorta in delayed phases by one radiologist with 5 years of experience.
SNR and CNR were calculated based on following standard equations, according to Bhosale P., et al. :
Subjective image quality
Subjective image quality and bladder carcinoma were initially assessed independently by two experienced radiologists with more than 20 and 12 years of experience in abdomino-pelvic CT performed visual assessments in blinded randomized fashion. The radiologists rated visual image noise and according to a three-point score (1, unacceptable; 2, average; 3, minimal), and overall image quality according to a 3-point score (1, unacceptable; 2, good; 3, excellent). The presence or absence of bladder lesion was assessed (0, absent; 1, present). Differences between two readers were solved by consensus. Results were then compared with cystoscopic examination results and pathological analysis.
Statistical analysis was performed with SPSS 20.0 (IBM Corp. Released 2013. IBM SPSS Statistics for Windows, Version 22.0. Armonk, NY: IBM Corp.).
Demographic patient characteristics
Detailed characteristics of patients were provided in Table 1. The 21 patients underwent standard-dose CT urography protocol A and additional limited scan to the pelvis at delayed phase protocol B. scan protocols were subdivided into 4 groups protocol A1 and B1 (patients ≤ 80 kg; 10 patients 47.6%) and protocol A2 and B2 (patients > 80 kg; 11 patients 52.3%).
The mean volume CT dose indices (CTDI) for group A1(130 kVp) and B1(80 kVp) were 6.64 ± .40 and 1.95 ± 1.12 mGy, respectively, while for groups A2(130 kVp) and B2(110 kVp), it was 6.71 ± .46 and 4.58 ± .35 mGy, respectively; this represents decrease by 71.5% for 80 kVp protocol (B1) and 31.7% for the 110 kVp protocol (B2) versus standard-dose protocols; this is described in details in the Table 2.
The mean DLP in protocol A1(130 kVp) was 323.91 mGy-cm ± 25.16 and for protocol B1 (80 kVp) 97.43 mGy-cm ± 54.25 which is 74.7% decrease and for protocol A2 (130kVp) 339.78 mGy-cm ± 25.70 and for protocol B2(110 kVp) 213.31 mGy-cm ± 38.52 which is 39.1% decrease as shown in Table 2. The mean ED for protocol A1(130kVp) was 5.49 mSv ± .43 and for protocol B1 (80 kVp)was 1.65 ± .92 with 70.8% decrease while for protocol A2(130 kVp) and B2 (110 kVp) mean ED was 5.77 mSv ± .43 and 3.62 mSv ± .65, respectively, with 38.1% decrease (Fig. 1).
Objective image quality assessment
Protocol A1 vs. Protocol B1: There were significant differences in SNRs between protocol A1 and B1 at aorta and psoas muscles.
Protocol A2 vs. Protocol B2: No significant differences were seen between protocol A2 and B2 (P ≤ 0.05 for all). The results of the objective image analyses are listed in Table 3.
There was no significant difference at CNR between different groups, as listed in Table 3.
Subjective image quality assessment
Protocol A1 vs. Protocol B1: There were significant differences between protocol A1(130kVp) and B1(80 kVp) as regards to visual image noise, and overall image quality as listed in (Table 3).
Protocol A2 vs. Protocol B2: There were no significant differences between protocol A2 (130 kVp) and B2 (110 kVp) as shown in Table 3.
Detection of bladder carcinoma
Diagnosis was identical among different protocols as regards per patient-level analyses among observers. Diagnostic accuracy of CT urography at standard and low-dose protocols was 91.3% (Fig. 2). Detailed results are listed in Table 4. Out of the 21 patients, 19 patients proved to have urothelial bladder cancer (90.4%); 1 of them had urothelial carcinoma with glandular and squamous cell differentiation which is considered bad prognostic factor. One patient had small cell carcinoma (4.7%) and One patient proved to be negative (chronic cystitis) as reported in CTU and proved by cystoscopy and biopsy result.
Two cases having multiple superficial small papillary lesions were reported as normal in both standard- and low-dose protocols CT while in cystoscopy, lesions were noted and biopsy results revealed urothelial bladder carcinomas. Per lesion diagnosis was the same for standard-dose and low-dose protocols. Diagnostic accuracy per lesion at standard-dose and low-dose protocols were 78.8%. Superficial small papillary lesions were missed in various cases at CT urography. Detailed results are listed in Table 4.
The radiation dose of CT urography is estimated to be approximately 15 to 20 mSv according to the ESUR guideline. The increasing dose awareness in the radiological and the general medical community has driven radiologists as well as equipment manufacturers to optimize CT imaging procedures with the aim of dose reduction .
In our study, we compare between standard dose CT with FBP and low-dose CT with IR at delayed phase only. By applying reduced-dose protocols in our study for detection of bladder cancer, CTDI and DLP and ED was reduced by average 72.3 % in the 80 kVp protocol (B1) and by 36.3% in 110 kVp (B2) protocol in comparison with standard-dose protocols, protocol A1 and protocol A2. There is significant decrease in radiation dose, the same as reported in previous studies [7,8,9,10].
In our study, attenuation and noise of tissues of interest were increased significantly when the tube voltage setting was lowered to 80 kVp, which resulted in lower SNR with 3.48 ± 1.56 at aorta and 2.23 ± 0.49 at psoas muscle in comparison with standard-dose protocol where SNR was 5.10 ± 1.64 at aorta and 3.72 ± 0.84 at psoas muscle. While in previous studies [7, 9], SNR of low-dose protocols were higher compared to standard-dose protocols. Difference between studies at noise levels and SNR at 80 kVp protocol could be attributed to variability in tube voltage and tube current. While there was no significant difference at CNR between standard-dose protocol (A1) and 80 kVp protocol (B1).
As regards to subjective image assessment, increased visual image noise degraded the image quality in 80 kVp, but did not affect the diagnostic accuracy, same as in Bahn et al. .
Whereas in 110 kVp, there were no significant differences were noted at SNR and CNR in comparison with standard-dose protocol A2. Subjective image analysis was similar to standard dose, for image noise and overall image quality. B2 protocol showed better lesion conspicuity and even a less image noise and better image quality than standard dose at 18.8% of cases for protocol B2 as reported by two experienced radiologists. This is consistent with Gervise et al.  and Bahn et al.  which confirmed that it is possible to perform an acquisition with 100 kV for most patients including overweight patients and obese patients.
Our results showed that the detection of urothelial carcinomas by the reduced-dose images with IR was the same as standard-dose with FBP. Lesion and patient-based accuracies were 78.72% and 91.30%, respectively, among different protocols that had the same range of patient body weight. As reported with Juri et al.  and Bahn et al. . The higher accuracies may be attributed to the larger tumor size. Superficial low-grade tumors are still challenging to detect regardless of standard-dose images with FBP or reduced-dose images with IR.
Limitations of the study
This includes small number of patients in this study. A study with larger number of patients is needed. Also, the study only assessed bladder carcinomas, and did not asses other related disease entities, including upper urinary tract abnormalities or lymphadenopathy. Moreover, we did not compare between different CT vendors.
This study showed the use of reduced-dose IR images was associated with significant reduction in radiation dose with no difference at diagnostic detection of bladder carcinomas compared with the use of standard dose FBP images, though IR was associated with an increased noise in 80 kVp images. These results may have important clinical implications using a low-tube voltage scan to reduce cumulative radiation exposure without compromising lesion detection.
Availability of data and materials
The datasets used and/or analyzed during the current study are available from the corresponding author on reasonable request.
CT dose index
Computed tomography urography
Filtered back projection
Multidetector computed tomography
Region of interest
Nawfel RD, Judy PF, Schleipman AR et al (2004) Patient radiation dose at CT urography and conventional urography. Radiology 232(1):126–132
Eikefjord EN, Thorsen F, Rørvik J (2007) Comparison of effective radiation doses in patients undergoing unenhanced MDCT and excretory urography for acute flank pain. Am J Roentgenol 188(4):934–939
Geyer LL, Schoepf UJ, Meinel FG et al (2015) State of the art: iterative CT reconstruction techniques. Radiology 276(2):339–357
Juri H, Matsuki M, Inada Y et al (2013) Low-dose computed tomographic urography using adaptive iterative dose reduction 3-dimensional: comparison with routine-dose computed tomography with filtered back projection. J Comput Assist Tomogr 37(3):426–431
Bhosale P (2015) Comparing CNR, SNR, and Image Quality of CT Images Reconstructed with Soft Kernel, Standard Kernel, and Standard Kernel plus ASIR 30% Techniques. Int J Radiol 2(2):60–65
Von Falck C, Bratanova V, Rodt et al (2013) Influence of sinogram affirmed iterative reconstruction of CT data on image noise characteristics and low-contrast detectability: an objective approach. PLoS One 8(2):e56875
Bahn YE, Kim SH, Kim MJ et al (2015) Detection of urothelial carcinoma: Comparison of reduced-dose iterative reconstruction with standard-dose filtered back projection. Radiology 279(2):471–480
Funama Y, Awai K, Nakayama Y et al (2005) Radiation dose reduction without degradation of low-contrast detectability at abdominal multisection CT with a low–tube voltage technique: phantom study. Radiology 237(3):905–910
Gervaise A, Naulet P, Beuret F et al (2014) Low-dose CT with automatic tube current modulation, adaptive statistical iterative reconstruction, and low tube voltage for the diagnosis of renal colic: impact of body mass index. Am J Roentgenol 202(3):553–560
Marin D, Nelson RC, Schindera ST et al (2010) Low-Tube-Voltage, High-Tube-Current Multidetector Abdominal CT: Improved Image Quality and Decreased Radiation Dose with Adaptive Statistical Iterative Reconstruction Algorithm—Initial Clinical Experience. Radiology 254(1):145–153
Ethics approval and consent to participate
• Done and the study was submitted to and approved by our ethics committee of Faculty of Medicine, Alexandria University.
• Ethics committee’s reference number: not available
• Formal written consent was taken from all participants
Consent for publication
Consent was taken from all participants.
The authors declare that they have no competing interests.
Springer Nature remains neutral with regard to jurisdictional claims in published maps and institutional affiliations.
About this article
Cite this article
Sharafeldeen, M., Shaaban, M., Ramadan, A.M. et al. Reduced dose iterative reconstruction versus standard dose filtered back projection in detection of bladder tumors. Egypt J Radiol Nucl Med 51, 100 (2020). https://doi.org/10.1186/s43055-020-00218-x
- Reduced dose
- CT urography
- Bladder mass